# Miniature Stereo-PIV 시스템의 개발과 응용

• Published : 2003.11.01

#### Abstract

Stereoscopic particle image velocimetry is a measurement technique to acquire three dimensional velocity field by two cameras. With a laser sheet illumination, the third velocity component can be deduced from out-of$.$plane velocity components using a stereoscopic matching method. Most industrial fluid flows are three dimensional turbulent flows, so it is necessary to use the stereoscopic PIV measurement method. However the existing stereoscopic PIV system seems hard to use since it is very expensive and complex. In this study we have developed a Miniature Stereo-PIV(MSPIV) system based on the concept of the Miniature PIV system which we have already developed. In this paper, we address the design and some primitive experimental results of the Miniature Stereo-PIV system. The Miniature Stereo-PIV system features relatively modest performances, but is considerably smaller, cheaper and easy to handle. The proposed Miniature Stereo-PIV system uses two one-chip-only CMOS cameras with digital output. Only two other chips are needed, one for a buffer memory and one for an interfacing logic that controls the system. Images are transferred to a personal computer (PC) via its standard parallel port. No extra hardware is required (in particular, no frame grabber board is needed).

#### References

1. Prasad, A. K. and Adrian, R. J., 1993, 'Stereoscopic Particle Image Velocimetry Applied to Liquid Flow,' Exp. In Fluids, 15, pp. 49-60 https://doi.org/10.1007/BF00195595
2. Chetelat, O. and Kim, K. c., 2002, 'Miniature Particle Image Velocimetry System with LED In-Line Illumination,' Meas. Sci. Technol., Vol. 13, No.7, pp. 1006-1013 https://doi.org/10.1088/0957-0233/13/7/306
3. Chatelat, O., Yoon, S. Y. and Kim, K. C., 2001, 'Design and Construction of a Miniature PIV(MPIV) System,' KSME International Journal, Vol. 15, No. 12, pp. 1775-1783
4. Chetelat, O. and Kim, K. c., 2002, 'Design of a Particle Image Velocimetry Objective Maximizing the Image Signal-to-Noise Ratio,' Meas. Sci. Technol., Vol. 13. No.5, pp. 667-682 https://doi.org/10.1088/0957-0233/13/5/304
5. Jacquot, P. and Rastogi, P. K., 1981, 'Influence of Out-of-Plane Deformation and Its Elimination in White Light Speckle Photography,' Opt. Lasers Eng., Vol. 2, pp. 33-55 https://doi.org/10.1016/0143-8166(81)90011-7
6. Prasad, A. K., 2000, 'Stereoscopic Particle Image Velocimetry,' Exp. In Fluids, 29, pp. 103-116 https://doi.org/10.1007/s003480000143
7. Prasad, A. J. and Jensen, K., 1995, 'Scheimpflug Stereocamera for Particle Image Velocimetry in Liquid Flows,' Appl. Opt., 34, pp. 7092-7099 https://doi.org/10.1364/AO.34.007092
8. Lawson, N. J. and Wu, J., 1997, 'Three-Dimensional Particle Image Velocimetry: Experimental Error Analysis of a Digital Angular Stereoscopic System,' Meas. Sci. Technol., Vol. 8, pp. 1455-1464 https://doi.org/10.1088/0957-0233/8/12/009
9. Soloff, S. M., R. J. Adrian, Liu, Z-c., 1997, 'Distortion Compensation for Generalized Stereoscopic Particle Image Velocimetry,' Meas. Sci. Technol., Vol. 8, pp. 1441-1454 https://doi.org/10.1088/0957-0233/8/12/008