- Volume 34 Issue 11
DOI QR Code
Numerical Analysis and Optimum Design of Disposable Drug Infuser Using Fluid-Structure Interaction Technique
유체-구조 상호작용기법을 이용한 일회용 약물주입기의 성능 해석 및 최적 설계
- Kim, Heon-Young (Dept. of Mechanical and Biomedical Engineering, College of Engineering, Kangwon National Univ.) ;
- Kim, Hak-Jin (Dept. of Mechanical and Biomedical Engineering, College of Engineering, Kangwon National Univ.)
- Received : 2010.03.12
- Accepted : 2010.09.14
- Published : 2010.11.01
A disposable drug infuser is used to provide drugs to patients who are not hospitalized; in this infuser, an elastic recovery force is exerted by a diaphragm made of a rubber-like materialsuch that a constant amount of drugs is provided to a patient. The drug infuser has to control the speed and amount of drugs to be released, as well as the overall duration for which they are to be administered. However, in a drug infuser with an elastic diaphragm, the infusion pressure depends on the amount of drug remaining within the infuser, and the amount of drug infused gradually decreases as the amount remaining in the infuser decreases. In this study, a finite element procedure involving the application of the fluid-structure interaction technique was developed and the performance of the elastic type disposable drug infuser was analyzed. The optimum design for ensuring that the infusion pressure remains constant throughout the duration of usage, including during infusion and discharge, was determined by this procedure.
Disposable Drug Infuser;Fluid-Structure Interaction;Taguchi Methods
- Skokal W. A., 1997, "Infusion Pump Update," RN 60, pp.35-38.
- Dorr R. T., Trinca C. E., Griffith K., Dombrowsky P. L. and Salmon S. E., 1979 "Limitations of a Portable Infusion Pump in Ambulatory Patients Receiving Continuous Infusions of Anticancer Drugs," Cancer Treat Rep, No.63, pp.211-213.
- Nam, S. B., Chang, C. H.. Rhee, H. D. and Lee, Y.-W., 2002, "Clinical Application of Gas Forming Infusion Pump for Intravenous Postoperative Pain Control," The Korean Pain Society, pp.190-197.
- Munson, B. R., Young, D. F. and Okiishi, T. H., 2006, "Fundamentals of Fluid Mechanics," pp.319-369.
- ADINA R&D, 2008, "Theory and Modeling Guide," Vol.1, pp.615-627.
- Degroote, J., Bathe, K.-J. and Vierendeels, J., 2008, "Performance of a New Partitioned Procedure Versus a Monolithic Procedure in Fluid–Structure Interaction," Computers and Structures, Vol.87, pp.793-801. https://doi.org/10.1016/j.compstruc.2008.11.013
- Li, C. and Lua, J., 2009, "A Hyper-Viscoelastic Constitutive Model for Polyurea," Materials Letters, Vol.63, pp.877-880. https://doi.org/10.1016/j.matlet.2009.01.055
- Ilfeld, B. M., Morey, T. E. and Enneking, K. F., 2003, "Portable Infusion Pumps Used for Continuous Regional Analgesia: Delivery Rate Accuracy and Consistency," Regional Anesthesia and Pain Medicine, Vol.28, No.5, pp.424-432.
- Ackermann, M., Maier, S., Ing, H. and Bonnabry, P., 2007, "Evaluation of the Design and Reliability of Three Elastomeric and One Mechanical Infusers," Journal of Oncology Pharmacy Practice Vol. 13, No.2, pp.77-84. https://doi.org/10.1177/1078155207078349