- Volume 9 Issue 3
DOI QR Code
Development and Performance Evaluation of Radial Exhaust Multi-port System for Real-time Particle Size Distribution Measurement
실시간 입자분포 측정을 위한 Radial Exhaust Multi-port System의 개발 및 성능평가
- Lee, Hong Ku (Department of Mechanical Engineering, Hanyang University) ;
- Lee, Yang-Woo (Department of Mechanical Engineering, Hanyang University) ;
- Jeon, Ki Soo (Department of Mechanical Engineering, Hanyang University) ;
- Ahn, Kang-Ho (Department of Mechanical Engineering, Hanyang University)
- Published : 2013.09.30
Measuring particle size distribution is one of the primary concerns in aerosol studies. For a nano-particle size distribution measurement, many scientists use a combination of a differential mobility analyzer (DMA) and a condensation particle counter (CPC) system, which is a called scanning mobility particle sizer (SMPS). Although it has a very high particle size resolution, some issues still remain. These problems include residence time between a DMA and a CPC, discontinuity of a CPC, and disturbance due to long scanning time during the precise measurement of particles. In particular, long scanning time is not adequate for measuring particle size distribution since the particle concentration is changing during the measurement. In this study, we developed radial exhaust multi-port system (REM-system) with no scanning time and high resolution to measure real-time particle size distribution. As a result of the REM-system performed using mono-disperse particle, it is expected that this system will be suitable for measuring continuously changing aerosol. If the counting efficiency of multi-condensation particle counter (M-CPC) and data inversion matrix are completed, REM-system will be a very adequate system for unsteady aerosol, which changes for SMPS scanning time.
- Ahn, K.H., H.N. Kim, J.H. Lee, G.N. Bae (1996). Particle Path and Performance Evaluation of Differential Mobility Analyzer, J. Mech. Sci. Tech., 20, 2005-2013.
- Chen, D.R., W. Li, M.D. Cheng (2007), Development of a Multiple-Stage Differential Mobility Analyzer(MDMA), Aerosol Sci. Tech., 41, 217-230. https://doi.org/10.1080/02786820601134551
- Fissan, H. J., Helsper, C., Thielsen, H. J. (1983). Determination of particle size distributions by means of an electrostatic classifier, J. Aerosol Sci., 14, 354-357. https://doi.org/10.1016/0021-8502(83)90133-7
- Kim, J.S., H. Chung, K.H. Ahn (2007). Development and Evaluation of Multi-Port System for Real-time Size Distribution Measurement, Proc. 2007 KAPAR conference, 69-70.
- Knutson, E. & Whitby, K., (1975). Aerosol Classification by Electrical Mobility : Apparatus, Theory and Applications, J. Aerosol Sci., l6, 443-451. https://doi.org/10.1016/0021-8502(75)90060-9
- Kousaka, Y., K. Okuyama and M. Adachi (1985). Determination of particle size distribution ultra- fine aerosols using a differential mobility analyzer, Aerosol Sci. Tech., 4, 209-225. https://doi.org/10.1080/02786828508959049
- Rader, D.J., P.H. McMurry (1986), Application of the tandem differential mobility analyzer to studies of droplet growth or evaporation, Aerosol Sci. Tech., 17, 771-787. https://doi.org/10.1016/0021-8502(86)90031-5
- Shih Chen Wang and Richard C. Flagan, and John H. Seinfeld (1990). Scanning electrical mobility spectrometer, Aersol Sci. and Tech., 13, 230-240. https://doi.org/10.1080/02786829008959441
- Wiedensohler, A., (1988). An approximation of the bipolar charge distribution for particles in the submicron size range, J. Aersol Sci.,19, 387-389. https://doi.org/10.1016/0021-8502(88)90278-9