- Volume 38 Issue 3
DOI QR Code
Fatigue Strength of Dental Implant in Simulated Body Environments and Suggestion for Enhancing Fatigue Life
생체유사환경 하의 치과용 임플란트의 피로강도 평가 및 수명 향상법
- Kim, Min Gun (Dept. of Mechanical & Biomedical Engineering, Kangwon Nat'l Univ.)
- 김민건 (강원대학교 기계의용공학과)
- Received : 2013.06.25
- Accepted : 2013.11.18
- Published : 2014.03.01
Fatigue tests were performed in various simulated body environments reflecting various factors (such as body fluids, artificial saliva) relevant within a living body. First, the fatigue limit under a simulated body environment (artificial saliva) was evaluated and the governing factors of implant fatigue strength were looked into by observing the fracture mode. The fatigue life of an implant decreased in the artificial saliva environment compared with that in the ringer environment. Furthermore, in the artificial saliva environment, the implant fracture mode was fatigue failure of fixture as opposed to the abutment screw mode in the ringer environment. In the fatigue test, corrosion products were observed on the implant in the simulated body environment. A larger amount of corrosion products were generated on the artificial saliva specimen than on the ringer specimen. It is thought that the stronger corrosion activity on the artificial saliva specimen as compared with that on the ringer specimen led to an overall decrease of fatigue life of the former specimen. In the case of the implant with a nitrided abutment screw eliminated hardened layer (TixN), a several times increase in fatigue life is achieved in comparison with tungsten carbide-coated implants.
Dental Implant;Simulated Body Environment;Artificial Saliva;Fatigue Life;Corrosion Fatigue;Corrosion Products
- Jemt, T. and Lekholm, U., 2003, "Oral Implant Treatment in Posterior Partially Edentulous Jaws: a 5-Year Follow-up Report," International Journal of Oral and Maxillofacial Implants, Vol. 8, pp. 635-640.
- Boggan, R. S., Strong, J. T., Misch, C. E. and Bidez, M. W., 2003, "Influence of Hex Geometry and Prosthetic Table Width on Static and Fatigue Strength of Dental Implants," The Journal of Prosthestic Dentistry, Vol. 82, No. 4, pp. 436-440.
- Son, J. H., Yang, Y. S. and Lee S. Y., 2005, "Fatigue Life Estimation of Implant Using a Finite Element Method," Trans. Korean Soc. Mech. Eng. A, No.22, pp. 998-101.
- Long, M. J. and Rack, H. J., 1998, "Titanium Alloys in Total Joint Replacement," Materials Science Perspective Biomaterials, Vol. 19, pp. 1621-1639.
- Esposito, M., Hirsch, J. M., Lekholm, U. and Thomsen, P., 1998 "Biological Factors Contributing to Failures of Osseointegrated Oral Implants (II) Etiopathogenesis," European Journal of Oral Sciences, Vol. 106, pp. 721-764. https://doi.org/10.1046/j.0909-8836..t01-6-.x
- Lee, B. J. and Kim, M. G., 2003, "An Electrochemical Approach to Development of a Method for Accelerated Strength Evaluation of Hard Tissue Replacement Materials," METALS AND MATERIALS International, Vol. 9, No. 2, pp. 215-219. https://doi.org/10.1007/BF03027280
- "Dentistry-Implants-Dynamic Fatigue Test for Endosseous Dental Implants," ISO 14801, The International Organization for Standardization. 2007.
- Lee, B. J., Jun, K., Masao, S., 1999, "Corrosion Fatigue Behavior of Ti-6Al-4V Ally in Simulated Physiological Envirnment," Biomaterial, Vol. 17, No. 1. pp.24-31.
- Kim, M. G., 2011, "Fatigue Properties on the Failure Mode of a Dental Implant in a Simulated Body Environment," METALS AND MATERIALS International 17-5, pp.705-711. https://doi.org/10.1007/s12540-011-1002-6
- Kim, M. G., 1999, "A Study of the Ruling Factor of Fatigue Limit of Nitrided Titanium Using Fluidized Bed Furnace," Trans. Korean Soc. Mech. Eng. A, No. 23, pp. 31-36.
- Adell, R., Erikkson, B., Lekholm, U., Branemark, P. I. and Jemt, T.,1990, J. Oral Maxillofac Implants Int. 5, pp.347-359.
- Morita, T., Kato, K., Shimizu, M. and Kawasaki, K., 1997, "Comparison of Fatigue Properties of Nitrided Pure Iron and Titanium," Trans. of the JSME(A), No.63-605, pp. 1-6.
- Effect of shot peening on the corrosive resistance of Ti-alloy under simulated body environment vol.42, pp.4, 2015, https://doi.org/10.14815/kjdm.2015.42.4.385